Cardiovascular Systems, Inc. (CSI®) (NASDAQ: CSII) has initiated a voluntary recall of unused WIRION Embolic Protection Systems (WIRION) due to complaints of filter breakage during retrieval.
CSI has informed all affected healthcare facilities to discontinue use of WIRION immediately and return unused product to CSI. The U.S. Food and Drug Administration (FDA) has also been notified. To date, CSI has received 9 complaints of filter breakage during retrieval. Filter breakage may result in device embolism and possible additional intervention.
The company plans to voluntarily recall all WIRION units currently in customer inventory. In total, 697 devices were distributed in the U.S. between March 22, 2021 and November 15, 2021. All lots of the following models are being recalled:
Name: WIRION Embolic Protection Device
Physicians and healthcare facilities can direct questions to their CSI representative or, call 651-259-2800. Adverse reactions or quality problems experienced with the use of the WIRION System may be reported to the FDA MedWatch Adverse Event Reporting program either online, by regular mail or by fax:
- Complete and submit the report Online: www.fda.gov/medwatch/report.htm